With the logic analyzer plugged into your computer, start the Logic 2 software. If this is the first time you are using your logic analyzer, the Logic 2 software will automatically download a calibration file for your particular analyzer. If you do not have access to the Internet, refer to the Device Calibration section of the User Guide.
Before you capture any data, click on the Capture Settings button to configure the logic analyzer. Select the sampling rate, capture duration, ports, voltage, and other settings relevant to your preferred capture parameters.
Note: only the Logic 8 Pro and Logic 16 Pro have the "Voltage" setting.
To learn more, see the Collecting Data and Device Settings section of the User Guide.