When measuring rise times and fall times, we recommend using your device's highest advertised sample rate.
Inside Logic 8, Logic Pro 8, and Logic Pro 16, there is a digital analog filter that engages for sample rates lower than the advertised sample rate. It is called the decimation filter and will filter the data further before down sampling so data sampled at lower rates does not suffer from aliasing from frequency components that made it through our high-order, anti-aliasing filter from the previous stage.
You can read more about Logic's filter stages here: Why Does My Signal Appear Correctly in the Digital Capture but Does Not Look Right or Is Missing Completely from the Analog Recording on the Same Channel?
This means that when measuring at lower sample rates, the rise and fall times may appear longer than they actually are due to our decimation filter. Because of this, rise and fall times should be measured at your device's highest advertised sample rate.
Tags: Saleae Logic software rounds off signal edges. Signal transitions seem slow. The recorded signal looks filtered. The edge steepness looks incorrect.
trello ID: 590bb7a70a7cb64b937fb7fa
trello data: -687750693